Measuring by HPLC a particular component of complex samples such as natural products, body fluids, environmental probes is complicated due to interference from other components. Different sample matrices require different sample cleaning procedures prior to separation and as a result, one procedure cannot be used across a variety of products. The column’s operating life is usually short, even with sample cleaning, due to irreversibly retained contaminants. Thus, long cleaning procedures, guard columns, and expensive and complex compound specific detection techniques such as MS are often required. An example of such problems appear in the measuring of nitrates. All of the existing methods are complicated and take significant time to execute. Usually, the methods are specific only for the particular sample type.
FlipLC™ is an alternative method to avoid the interference of most of the contaminants by the use of an isolation column and a high pressure switching valve before the separation column. This method allows sample cleaning and analyte separation in one automated process.The isolation column and the separation column should have orthogonal retention characteristics to operate efficiently in this setup. Mixed-mode columns with reverse phase and ion-exchange characteristics were used in this analysis. The setting below employs a high pressure 6-port, 2-position switching valve which is placed between an autosampler and analytical column and actuated by the chromatography system by a defined time after the injection.
An additional isolation column is connected to the valve’s ports and allows the switch of the position of this column in the chromatography flow path. In this setup, a sample is introduced to the isolation column first. The isolation column passes analytes through and retains the late eluted components of the sample. When the valve is actuated at a defined time after the injection (this time depends on the nature of the analyte) the whole plumbing system of the setting changed without flow interruption. After the valve switching (takes fraction of a second) the isolation column flip in reverse relative to the mobile phase flow and become connected to the detector outlet. The late eluted components, while still in the isolation column, are back-flushed to waste by the flow coming from the detector outlet. The original plumbing arrangement restored prior to next injection.
Many biologically important compounds can be analysed directly in human serum/plasma. Before, the restrictive access media (RAM) columns was used which traps small molecules on RAM column while plasma proteins do not retain. Then small molecules can be analysed on regular analytical RP column with UV or MS detector. However this technique does not work for polar molecules. They do not retain on RAM columns with hydrophobic interior. This became easy task for FlipLC. The serum proteins get trapped on flip Primesep SB column at low organic mobile phase and polar analytes pass through directly to analytical column. After valve switching the proteins get flushed out in reverse direction from Flip column by a secondary pump using solvent with higher organic concentration. In Mass Spec (MS) detection setting the mobile phase gets evaporated in the detector and in order to flush the Flip column an additional isocratic pump would required (see diagram below).
As a result, a good isolated peak of polar analyte was obtained with serum proteins removed from the sample. Because of the efficiency of the analyte isolation, the final method can be done in about 5 minutes.
One useful application presented is analysis of serotonin in human serum. This setting eliminates the need for complex sample cleaning prior to the analysis, provide high analytes recovery, eliminate needs for SPE cartridges and liquid handling equipment. The method is fully compatible with MS and UV detection.
These results were obtained due to the unique properties of mixed-mode columns, which retain compounds by both reverse phase (RP) and ion-exchange (IE) mechanisms.
FlipLC has been used for measuring different charged compounds in complex mixtures.
The list includes:
If the sample is a complicated mixture consisting of many different chemicals, it is hard to isolate a particular analyte without complicated sample preparation. Our new FlipLC method addresses this problem. It consists of an analytical column connected by a valve to an isolation (flip) column which retains charged analytes orthogonally relative to the analytical column. This method eliminates most of the interfering components from the chromatogram, shortens the analysis time and increases the lifetime of the analytical columns. FlipLCTM allows for the analysis of a targeted compound found within a variety of complex samples using a single chromatography method. Due to unique property of mixed-mode columns to retain molecules by RP and IE mechanizm, many polar charged analytes can retain and quantitate with simple methods without ion-pairing reagents.